The scope, as it arrived. The previous owner has painted the enclosure; unfortunately the lettering on the back panel was painted over and the color is a bit too 'blue' to match the original Tek color. Other than that, it's in excellent physical condition. The waveform appears slightly blurred because the scope does have a triggering issue (meaning that it doesn't), so the photo was taken with the function generator set to give as stable a waveform as possible. it managed to shift a bit during the almost half second exposure when the photo was taken.
Tektronix 516Tek 516dual trace scope